A kind of lower limb rehabilitation training device
This utility model is related to medical treatment and nursing apparatus technical field, more particularly, to a kind of lower limb rehabilitation training device.
Paralysis is a kind of common frequently-occurring disease, and the side limbs of patient are stiff, bring greatly to patient and household
Misery.Treatment hemiplegia method except take medicine in addition it is also necessary to routing motion take exercise, to strengthen blood circulation.Especially ill first
In three months of phase, it is crucial convalescence.In this stage, such as carry out lasting motion not in time, can be formed joint stiffness,
The phenomenons such as amyotrophy, the rehabilitation after giving causes difficulty.High with artificial nurse fees, and because being affected by multiple anthropic factors,
Nursing is difficult to continue long-term effect, affects nursing quality, hinders later stage rehabilitation.
Medical treatment and nursing apparatus can help people to carry out rehabilitation training, and with scientific and technical constantly progressive and development,
Increasing medical treatment and nursing apparatus occurs in medical instruments field, existing people carry out lower limb body Post operation/under
The handicapped patient of limb, needs the foot to patient and knee joint to carry out reconditioning；At present, existing rehabilitation training mode
Generally：Go to be trained in specific medical space, which is not only relatively costly, and need accompanying of household, to trouble
The family members of person bring inconvenience, in addition, which can not carry out rehabilitation training in real time, so can cause the optimal exercise of patient
Opportunity.And, the medical treatment and nursing apparatus being used for rehabilitation training in prior art designs not reasonable it is impossible to effectively help patient
Carry out rehabilitation training, single function, have to someone during patients ' recovery training and accompany, do not only take up the plenty of time, and recover effect
Really undesirable, and for some rehabilitation centers, fee charged is expensive, needs huge economic support.
Utility model content
Not enough for prior art, this utility model provides a kind of lower limb rehabilitation training device.
For achieving the above object, this utility model provides following technical scheme：
A kind of lower limb rehabilitation training device, including rack body, foot-operated body, contiguous block, described lower limb rehabilitation training dress
Put and also include chair, described chair includes chair face, chair back, armrests, described chair is arranged on chair bases top, institute
Stating chair bases with described rack body is to be fixedly connected, the upper surface of described chair bases and described rack body junction
It is provided with motor, described rack body is crossed-over by some pack supports, and be provided with the middle part of rack body
Lifting column, the top of lifting column is provided with contiguous block, and described contiguous block middle setting is through-hole structure, and runs through in through hole and set
Put rotating shaft, in the two ends of rotating shaft connection leg respectively, described supporting leg is connected with foot-operated body, the surface of foot-operated body is provided with
Some through holes, and the bottom of through-hole structure is provided with elastic reset convex body.
Preferably, the bottom of described rack body is provided with some anti-skidding cushion blocks.
Preferably, described elastic reset convex body is made up of the back-moving spring of bottom and the convex body at top.
Preferably, described foot-operated body side is hinged is provided with bandage.
Preferably, described armrests is provided with speed and adjusts button and switch.
Preferably, described motor connects described speed regulation button, switch and rotating shaft respectively by wire.
A kind of lower limb rehabilitation training device that this utility model provides, this utility model has the advantages that：
1st, assist operation in patients metapedes dorsiflexion plantar flexion and knee joint bends and stretches initiative exercise it is not necessary to other people assist；2nd, this forging
Smelting method not only has science, and patient can adjust button with Negotiation speed and voluntarily control exercise frequency, thus exciting actively
Temper enthusiasm, promote suffering limb function early stage to recover comprehensively；3rd, pass through the foot pedal structure of design on device, trouble can be significantly facilitated
Person carries out the Exercises of foot, the symptom of the patient of effective mitigation capability obstacle, and foot-operated elastic reset convex body structure is also
The effect of massage can be effectively played, promote the blood circulation of foot, whole device is reasonable in design, simple to operate, can be further
Popularization and application；4th, low cost, using comfortable, can be used as family care aid after patient discharge；5th, range is extensive,
For the patient of lower limb disorder, such as lower limb fracture surgery is in convalescence or because Other diseases lead to the trouble of lower extremity paralysiss
Person, all can be suitable for.
In order to be illustrated more clearly that this utility model embodiment or technical scheme of the prior art, below will be to embodiment
Or in description of the prior art the accompanying drawing of required use be briefly described it should be apparent that, for ordinary skill people
For member, on the premise of not paying creative work, other accompanying drawings can also be obtained according to these accompanying drawings.
The structural representation of Fig. 1 this utility model embodiment；
The overlooking the structure diagram of Fig. 2 this utility model embodiment；
The foot-operated body side structure schematic diagram of Fig. 3 this utility model embodiment.
In figure：1st, rack body；2nd, ride body；3rd, contiguous block；4th, lifting column；5th, chair face；6th, chair back；7th, chair
Handrail；8th, speed adjusts button；9th, switch；10th, chair bases；11st, motor；101st, anti-skidding cushion block；21st, supporting leg；22nd, through hole；
23rd, convex body；24th, back-moving spring；25th, bandage；31st, rotating shaft.
In order that the purpose of this utility model, technical scheme and advantage become more apparent, with reference to embodiments, to this
Utility model is further elaborated.It should be appreciated that specific embodiment described herein is only in order to explain this practicality
New, it is not used to limit this utility model.
Below in conjunction with the accompanying drawings and specific embodiment is further described to application principle of the present utility model.
As shown in Figure 1, Figure 2 and Figure 3, the present embodiment provides a kind of lower limb rehabilitation training device, including rack body 1, foot
Step on body 2, contiguous block 3, lower limb rehabilitation training device also includes chair, chair includes chair face 5, chair back 6, armrests 7,
Chair is arranged on chair bases 10 top, and chair bases 10 are to be fixedly connected with rack body 1, chair bases 10 and rack body
The upper surface of 1 junction is provided with motor 11, and rack body 1 is crossed-over by some pack supports, and in rack body 1
Middle part is provided with lifting column 4, and the top of lifting column 4 is provided with contiguous block 3, and contiguous block 3 middle setting is through hole 22 structure,
And run through setting rotating shaft 31 in through hole 22, in the two ends of rotating shaft 31 connection leg 21 respectively, supporting leg 21 is connected with and rides this
Body 2, the surface of foot-operated body 2 is provided with some through holes 22, and the bottom of through hole 22 structure is provided with elastic reset convex body 23.
Foot-operated body 2 rotates around rotating shaft 31, reaches effective effect tempering foot's limbs.Foot by design on device
Step on body 2 structure, in conjunction with the design of chair, patient voluntarily takes exercise conveniently, the activity instruction that patient carries out foot can be significantly facilitated
Practice, the symptom of the patient of effective mitigation capability obstacle,
The bottom of rack body 1 is provided with some anti-skidding cushion blocks 101, plays the effect of stable whole device.
Elastic reset convex body is made up of the back-moving spring 24 of bottom and the convex body 3 at top, and convex body 3 passes through through hole 22 according to making
With foot's behaviour in service Ou Ang of patient so that convex body 3 can with sole position abundant structure, the effect of massage significantly, play rush
Enter relatively low blood circulation.
The hinged effect being provided with bandage 25, playing fixed foot portion of foot-operated body 2 side, patient easy to use carries out foot
Restorative motion, whole device is compact and flexible, easy to utilize.
Speed is provided with armrests 7 and adjusts button 8 and switch 9, connection speed is adjusted motor 11 respectively by wire
Button 8, switch 9 and rotating shaft 31, speed adjusts button 8 and switch 9 is arranged on armrests 7, contributes to patient oneself at any time
Control beginning and the closing of foot-operated body 2, patient can adjust, with Negotiation speed, the frequency that button voluntarily controls foot-operated body 2,
Thus exciting initiative exercise enthusiasm, promote suffering limb function early stage to recover comprehensively.
Above example only in order to the technical solution of the utility model to be described, is not intended to limit；Although with reference to aforementioned reality
Apply example this utility model has been described in detail, it will be understood by those within the art that：It still can be to front
State the technical scheme described in each embodiment to modify, or equivalent is carried out to wherein some technical characteristics；And these
Modification or replacement, do not make the essence of appropriate technical solution depart from spirit and the model of this utility model each embodiment technical scheme