A kind of ventilative water bed mattress of mitigation deadweight
This utility model belongs to emergency medical technique with the apparatus field, is related to a kind of ventilative water bed mattress of mitigation deadweight.
, in medical field, for the patient due to outside bang or visceral injury, need by way of decompression
Lain low, water bed mattress device pliability is suitable for quickly lying low, but water bed mattress filled water amount is more, and deadweight is larger, is unfavorable for after water-filling
Move and move, it is more painstaking during hospital bed promotion also to lead to, and is unfavorable for using simultaneously；Because hospital bed needs piecemeal to deform,
So the mattress that present medical bedstead is used, the splicing mattress of typically all common material, the comfortableness that these mattresses have compares
Difference, is unfavorable for the recovery of patient, and for the user of long-time bed, on the one hand long-time bed is more tired, and experience is poor,
Simultaneously for the patient that can not take care of oneself, easily produce pressure ulcer；As using air cushion solve pressure ulcer, but air cushion patient is not had more
The quick restitution of auxiliary it is impossible to significantly mitigate nursing work pressure.
Meanwhile, current water bed mattress does not possess magnetic therapy functions and heating function and lives it is impossible to meet some simply fast quick-recoveries
Dynamic, easily cause to patient and constrain for a long time.
Content of the invention
For the problems referred to above, the technical problems to be solved in the utility model is the water bed providing a kind of mitigation deadweight ventilative
A kind of ventilative water bed mattress of mitigation deadweight of the present utility model, including bed body and scalable temperature controller；Described bed body
Surface is divided into multiple regions by reinforcement；Described bed body side is provided with water-filling steam vent, and opposite side is provided with current valve
Steam vent；Described bed body bottom is provided with discharge outlet；The inside bottom side of described bed body embeds EPE pearl wool or the soft pad of occupy-place
Plate；The bottom of described bed body is provided with back hanging bag；It is provided with permanent magnetism piece, permanent magnetism piece is arranged on plastic strip heat in described bed body
In the cloth bag of synthesis；Described water-filling aerofluxuss in the hole is provided with miniature pore water pressure sensor；It is additionally provided with use in described bed body
Electric heating wire in bed body heating；Described electric heating wire is electrically connected with scalable temperature controller.
Further, described electric heating wire adopts two-wire parallel way, is added using 9.6 meters of long carbon fibers of 2 24K every
Hot line composes in parallel.
Further, described occupy-place is fixed on bottom side in bed body with soft backing plate heat seal, and occupy-place is with leaving on soft backing plate
Further, described water-filling air vent is provided with the anti-spring of exhaust cap.
Further, described bed body top is provided with sink for head washing, feet washing trough and discharge port.
Further, described bed body is provided with the first passage of up/down perforation.
This utility model beneficial effect：This utility model substantially reduces the deadweight of bed body by occupy-place implant, is beneficial to
Move and move, be provided simultaneously with magnetic therapy functions, improve and microcirculation in human body is got through with efficiency, function of temperature control can preferably increase doctor
Treat the comfort level of experience.
For ease of explanation, this utility model is embodied as and accompanying drawing is described in detail by following.
Fig. 1 is the top plan of this utility model water bed mattress.
Fig. 2 is the bottom plan view of this utility model water bed mattress.
Fig. 3 is the inside electric heating wire winding diagram of this utility model water bed mattress.
Fig. 4 is the inner permanent magnetic piece scattergram of this utility model water bed mattress.
Fig. 5 is that the inside bottom side of this utility model water bed mattress embeds the implant schematic diagram for occupy-place.
The water bed mattress that a kind of mitigation deadweight as Figure 1-Figure 5 is breathed freely, including bed body 1 and scalable temperature controller；Described
Bed body 1 is made using flexible PVC material, and bed body 1 surface is divided into multiple regions by reinforcement 2, after water-filling in water bed mattress,
Using the natural flow of water and the flexible characteristic of PVC so that this water bed mattress Free Transform；
Described bed body 1 side is provided with water-filling steam vent 3, for aerofluxuss and water-filling；Described water-filling air vent 3 row of being provided with
The anti-spring of gas lid 4, for pushing down the exhaust cap on water-filling air vent, prevents exhaust cap from upspringing, leads to the generation of spilling water accident；Institute
State the anti-spring of exhaust cap 4 and there is elasticity, before opening exhaust cap, first pushed with handling the anti-spring of exhaust cap 4, Ran Houzai
Extract exhaust cap, press exhaust cap after being finished, anti-for exhaust cap spring 4 is restored to the original state, pushes down exhaust cap, described
The width of the anti-spring of exhaust cap 4 is identical with exhaust cap diameter width or halves.
The increase that the opposite side of described bed body 1 is provided with 1 or 2 valve outlet 11 valve outlet 11 against the current against the current saves
Labor strength, improves installation effectiveness, often installs a set of water bed mattress and can save 10～20 minutes about, and the set-up time will
Can shorten to 30 minutes about from original 50 minutes, substantially increase exhaust efficiency.
The bottom of described bed body 1 is provided with discharge outlet 8, back hanging bag 10, and described back hanging bag 10, for by bed body firmly
Be articulated on bed board, after hospital bed deformation is turned up will not run-off the straight, it is ensured that normal use the problems such as curling；Described draining
Mouth 8 is provided with discharge outlet opening regulating valve, can pass through the aperture area of discharge outlet opening governor valve control discharge outlet 8；One people
Just can be with complete independently draining it is not necessary to helper, user oneself can complete independently.
The inside bottom side of described bed body 1 embeds the soft backing plate 14 of EPE pearl wool or occupy-place；It is provided with described bed body 1
Permanent magnetism piece 13, permanent magnetism piece 13 is arranged in the cloth bag with plastic strip thermal synthesis；It is provided with miniature hole in described water-filling steam vent 3
Water pressure sensor；It is additionally provided with the electric heating wire 12 for bed body heating in described bed body 1；Described electric heating wire 12 with adjustable
Section temperature controller electrical connection；Popped one's head in by external warming and gather the temperature of bed body 1 diverse location, be transferred to scalable temperature controller, can
Adjust temperature controller and according to temperature change signal automatic and stop heating work；When the temperature difference of bed body 1 diverse location is big, plus
Big heating power, when the temperature difference hour of bed body 1 diverse location, reduces heating power.
Further, described electric heating wire 12 adopts two-wire parallel way, using 9.6 meters of long carbon fibers of 2 24K every
Heater wire composes in parallel.
Further, described occupy-place is fixed on bottom side in bed body 1 with the heat seal of soft backing plate 14, and occupy-place is with soft backing plate 14
Leave the second passage 15；Increase occupy-place in bed body 1 bottom side with soft backing plate 14, and with the material conduct with bed body 1 same material
Anticreep low water, occupy-place is fixed on bed body 1 bottom side with soft backing plate 14, and this material can use the resistance to compression such as XPE, EPE, EVA, resist
Deformation and light material that is ageing-resistant and not absorbing water, thickness can be 3～5cm；Meanwhile, occupy-place is led to leaving second on soft backing plate
Pore 15, in order to strengthen the breathability of water bed mattress.
Further, described bed body 1 is provided with the first passage 9 of up/down perforation, can strengthen the breathability of mattress.
Described bed body 1 top is provided with sink for head washing 5, feet washing trough 6 and discharge port 7, and described bed body 1 is run through in described discharge port 7
Top and bottom, thus providing the drain passage of Excreta；The concrete opening of described sink for head washing 5, feet washing trough 6 and discharge port 7
Position can be selected as needed.
Embodiment described above is only preferred implementation of the present utility model to be described, not to this practicality
New spirit and scope are defined.On the premise of without departing from this utility model design concept, this area ordinary person couple
Various modifications and improvement that the technical solution of the utility model is made, all should drop into protection domain of the present utility model, this reality
With the new technology contents being claimed, all record in detail in the claims.